|Type:||Medical Gas Equipment||Product Name:||Digital Switched Medical Gas Manifold|
|Instrument Classification:||Class II||Material:||Acrylic Panel, Metal Body|
|Color:||White Or Customized||Inlet Pressure:||2MPa ~ 15MPa|
|Outlet Pressure:||0.2MPa ~ 0.3MPa ( Oxygen Inhalation) 0.4MPa ~ 0.5MPa ( Ventilator Operation)||Delivery Flow:||20L/min ~ 60L/min ( Oxygen Inhalation) >=70L/min ( Ventilator Operation)|
|Inlet Thread:||G5/8"||Outlet Connection:||Φ8mm/M14*1.5 (mm)|
|Appearance Dimension:||376*200*100(mm)||National Utility Model Patent:||ZL 2018 2 0836558.7|
|Design Patent:||ZL 2018 3 0496789.3||OEM:||Available|
|Warranty:||24 Months||Package:||Carton Box Or As Your Order|
XCEL 2MPa Medical Gas Manifold,
XCEL 15MPa Medical Gas Manifold,
XCEL 20 bar medical oxygen manifold
The Digital Medical Gas Manifold can connect multiple Oxygen cylinders at the same time. It aims to reduce the high pressure of the Oxygen cylinders to a certain working pressure before Ventilator start up or Oxygen inhaling. It is equipped with a check-valve between each gas cylinder and the same Manifold respectively, which can help to control the high-pressure pipelines’ gas and to prevent the high-pressure gas cylinders’ gas to flow to the low-pressure gas cylinders.
There are two pressure regulators in the Digital Medical Gas Manifold, the gas pressure and flow for Ventilator and Oxygen Inhaling are preset respectively. Controlled by a switch changeover valve, the role is defined in three directions, Ventilator, OFF and Oxygen Inhaling. When the control Knob is switched to “OFF”, there are no output on the two low-pressure gas outlet connectors; when it is switched to the “Oxygen Inhaling”, the corresponding oxygen inhaling outlet connector has oxygen output, but the ventilator outlet connector has no oxygen output; vice versa.
The gas outlet connector is connected with the corresponding safety valve and LCD screen. The LCD screen is used to monitor the pressure in the high and the low pressure pipes of the manifold, and to display the capacity of the gas cylinder. When the capacity of the gas cylinder is lower than the set value, there is a flashing alarm function. The safety valve automatically exhausts when the output pressure is higher than the max pressure to ensure the flow’s safety and stability.
Cylinder’s pressure value and the remaining capacity
1. The pipeline is made of high-quality copper.The pipeline system is safe and reliable after pressure test.
2. Sliver brazing on piping joints for maximum leak prevention ;
3. Equipped with overpressure exaust device to ensure safe use ;
4. Equipped with a gas filter device to effectively block impurities in the gas ;
5. Control knob can be switched from "Ventilator" to "OFF",and from "OFF" to "Oxygen Inhaling";
Preset outlets pressure for different use and to be avoid of human intervention error in use;
More quickly, convenient and safe.
Model No. : X09105
|Technical Specification||Inlet pressure||2MPa～15MPa|
|Delivery Pressure||0.2MPa～0.3MPa(Oxygen Inhaling)|
|0.4MPa～0.5MPa( Ventilator Start-up)|
|Delivery Flow||20L/min～60L/min ( Oxygen Inhaling )|
|≥70L/min( Ventilator Start-up)|
National utility model patent: ZL 2018 2 0836558.7
Design patent: ZL 2018 3 0496789.3
Appearance Dimension: 376*200*100(mm)